Exploring surface sensitive technologies, you often encounter the real-time method of QCM. This three-letter abbreviation is short for Quartz Crystal Microbalance, which essentially means that this is a balance for very small masses. If you have browsed the instrumentation area a bit more, you may have noticed that there are several versions of QCMs, such as QCM-D. So, what is the difference between QCM and the four-letter version where a D is added at the end?Unlike traditional QCM, the QCM with an added D measures an additional parameter, namely the dissipation factor. The dissipation factor provides real time information about the softness and rigidity, also called the viscoelasticity, of the molecular layers on the sensor surface. This means that QCM-D provides more information about the system under study than traditional QCM.
So, QCM-D measures dissipation, but what is this parameter, what does it consist of and what does it mean? Sometimes dissipation is also referred to as damping of the oscillating sensor. The strict definition of dissipation is the sum of all energy losses in the system per oscillation cycle. Using an equation to describe it, it can be written as 1/Q, where Q is the so-called quality factor of the oscillator used. Another way to define dissipation is the energy dissipated per oscillation, divided by the total energy stored in the system.
So much for the definition, but how is this related to an actual experiment? As already mentioned, the dissipation correlates to the softness of the layers on the sensor surface. A rigid material deposited on the sensor surface will follow the sensor oscillation without being deformed during the oscillatory motion. This means that a rigid material will not dampen the oscillation and in this situation, the measured dissipation will be low. On the other hand, a soft film will not be able to follow the sensor oscillations entirely. This means that it will be deformed during the measurement. The soft film will dampen the oscillation and result in high dissipation. Consequently, the dissipation parameter offers a real-time characterization of the softness or viscoelasticity of films deposited on the sensor surface, and it makes it possible to monitor conformational changes.
Why is it important to measure the energy dissipation?
The possibility to collect more information about the system under study is of course positive, but is the dissipation information really needed? There are three important aspects of the dissipation parameter:
Going into more detail about the first important aspect, the qualitative insight of the time resolved behavior of the system under study, the dissipation parameter will complement the information from the collected resonance frequency. Together the two captured parameters f and D will paint a picture of how molecules arrange themselves on the surface and how the arrangement varies with time; the frequency response of a QCM sensor reflects the amount of mass coupled to the surface, including the mass of solvent which is trapped in the molecular layer.
The dissipation reflects the softness of the film. Monitoring these two parameters as a function of time makes it possible to detect for example if molecules are lying flat on the surface (rigid film and low level of hydration) or if they are in a stretched configuration (soft film and high level of hydration), as well as if there are rearrangements such as for example swelling (from flat to stretched) or collapse (from stretched to flat).
To summarize, the dissipation parameter offers three important benefits over traditional QCM.
To get more out of your experiments, you can use the information that the dissipation parameter provides.
Read more about the differences between QCM-D and other QCMs in our white paper.
This post was originally published in Nov 23, ’17.
One of the key factors to achieve reliable and reproducible QCM measurements is temperature stability. Here we list the top four factors that will help you eliminate temperature induced artifacts in your QCM data.
On the topic of QCM, the different resonance harmonics, i.e., the fundamental and the overtone harmonics, are often subject to discussion. So, what is the deal with the overtones? Are they really needed, and if they are, when and why are they important?
When you are to invest in a new QCM-system, there are several aspects of the instrument to evaluate such as price, experimental capabilities and data quality. Price is straightforward, but how about the other two factors? Here we guide you how to asses experimental capabilities and data quality.
You have finished a set of successful QCM experiments, and it is time to analyze the data. Mass and thickness quantification are on the to-do list, and you must decide between the Sauerbrey equation and viscoelastic modeling. Here we present guidelines on how to evaluate which method to use.
Temperature stability is key to achieve reliable and reproducible QCM measurements. But why is a stable temperature so important? And how will temperature variations affect your measurements? Here we explain the causes and effects of temperature related artifacts in QCM measurements.
Get updates from the blog directly to your inbox.